This is a solid state meter that measures AC, DC+, DC- Volts and Ohms. It has a mirrored analog meter. It's 1/3 Rack wide & will mount in 4 Units of height. The black common 5-way binding post is isolated from the instrument chassis. The AC measurements are average responding not true RMS.
I think the reason for the DC+ and DC- functions is becasue this meter reads up to 1,000 Volts on the DC range but may have a lower spec for the isolation between the black common 5-way binding post and earth ground. Some power supplies, like are used for small Cathode Ray Tubes (CRT) have the positive terminal connected to the chassis and the negative terminal is floated. The 427A could measure these voltages either direction.
The construction is discrete components on double sided printed circuit boards with plated through holes. The PCB on the right side looks as if it should be plugged into a socket (it has the protruding board with the traces to act as contacts) but instead there are individual wires with sockets on each that are plugged into pins. My guess is that (1) the PCB form factor was used on another high volume production instrument and that using the same form factor was a cost savings or (2) testing the board was done using a socket but the actual 427A meter used a higher reliability Gold to Gold contact arrangement for reliability.
Three of the Radio Shack 270-391 battery holders (4 ea. "AA") can be used to make a 16 cell "AA" battery adapter. When the rear panel switch is in the Line position and the FUNCTION switch is in the BATT (15V min) position, the meter reads 23.2 Volts on the 30 Volt Full Scale range. 16 Alkaline "AA" cells will provide 24 Volts and 16 NiCad or NMH will provide 19.2 Volts.
As of 29 Aug 2002 Eveready
still offers the 763 as a current product. It's a Zinc
Chloride 22.5 Volt battery with 1.7 Amp Hour capacity.
Radio Shack lists the battery as 960-0448 (NEDA 710).
I've read that a couple of common 9 Volt batteries connected in
series (18 V) will power the meter for a short amount of time.
00427-90004 (Jan 70) Operation & Service Manual (free from Agilent, BAMA), serial prefixed 947- [the foldouts are on separate pages]
00427-90005 for serial prefixes 560-, 621- & 731-
00427-90008 Serial number >= 0947A16701
Back to Brooke's Products for Sale, Military Test Equipment, Microwave Test equipment, Military Information, Home Alphanumeric index of web pages
Page created 29 Aug 2002.